Nickel Thin Film RTD Elements are characterized by a large change in resistance vs. temperature, and availability in several industry-standard temperature coefficients. We offer a variety of Nickel RTD sensors (Ni RTD) with different resistance values (Ni100, Ni120, Ni500, Ni1000) and different temperature coefficients (TC6180, TC 6720, TC 6370).

### Technical Data

Thin Film Nickel Temperature Sensors are configured as follows:

• Temperature Coefficient (TC) typically 6180 ppm
• Temperature Range from -260°C to +250°C
• Characterized by a large change in resistance vs. temperature, and availability in several industry-standard temperature coefficients

Product Size (L x W x H) Resistance Adicional Information
NI100 3.0 x 2.2 x 1.4 mm 100 Ω Temperature range -60 to 200°C
NI120 5.0 x 2.0 x 1.4 mm 120 Ω Temperature Coeficient 6720 ppm
NI500 5.0 x 2.0 x 1.4 mm 500 Ω
NI1000 5.0 x 2.0 x 1.4 mm 1000 Ω
NI1000 5.0 x 2.0 x 1.4 mm 1000 Ω @ 70°F (21.1 °C) Temperature Coeficient 6370 ppm
##### Equation of a nickel resistor according to DIN 43760:

$\fn_phv&space;\small&space;\fn_phv&space;\small&space;\fn_phv&space;\small&space;\fn_phv&space;\small&space;R_t=R_0&space;(1+At+Bt^{2}+Ct^{4}+Dt^{6})\Omega$

Where:

A=5,481 x 10-3

B=6,650 x 10-6

C=2,805×10-11

D =2,000×10-17

##### Maximum allowed tolerance as a function of temperature  (DIN 43760):

t<0°C: F = ±(0,4 + 0,028 x t) °C
t>0°C: F = ±(0,4 + 0,007 x t) °C